microdot® Code 69 Test Strips are pre-coded to work the the microdot® Code 69 Meter. This means you no longer have to code the meter each time you open a new vial of strips!
Glucometer Chemical Composition
Each microdot® strip contains the enzyme glucose dehydrogenase (Bacillus Sp.) = 1 IU and other ingredients (mediator, NAD, lysing agents etc.) = 200 µg.
When blood is added to the microdot® strip, the chemicals within the strip react to create a tiny electrical current. Your meter reads the current and displays your level of blood glucose.
The performance of microdot® Test Strips has been evaluated both in laboratory and in clinical tests.
The measurement range of the microdot® System is 20 to 525 mg/dL.
The accuracy of the microdot® System was assessed by comparing blood glucose results obtained by 102 patients at one clinical center with those obtained using a YSI 2300 laboratory Instrument.
Clark error grid analysis:
98% A region;
2 % B region
Correlation coefficient R = 0.977.
Range tested 45 to 472 mg/dL (2.5 to 26.2 mmol/L).
Slope 0.973y-Intercept 3.96 mg/dL (0.22 mmol/L).
This means that the microdot® system compared well to a laboratory method.